LANGAVANT CALORIMETER FOR HEAT OF HYDRATION OF CEMENT
Langavant calorimeter for heat of hydration of cement method consists of introducing a fresh cement specimen into a isolated Dewar flask and monitoring the temperature changes within the specimen during the first early days.
After a certain time, the heat of hydration of the cement content in the sample is equal to the sum of the heat accumulated in the flask and the heat emitted to the environment during the test period.
The temperature of the mortar is compared with the temperature of a inert sample placed in a reference calorimeter flask .
The amount of heat achieved by the cement mortar is mainly dependent on the nature thereof, and may reach values between 10 ° C and 50 ° C.
The amount of heat is expressed in joules per gram of cement.
It consist of 2 isolated calorimeter bottles set 2 temperature probes type PT-100 set with 3 threads, 50 disposable mortar box set and an electronics console with 4 measuring channels, and RS232 output interface for connecting up to 4 calorimeter bottles to the PC.
CM 0126 Langavant calorimeter for heat of hydration of cement complete set
CM 0126-1 Set of 2 isolated calorimeter bottles.
CM 0126-2 Set of 2 temperature probes type Pt-100, with 3 threads.
CM 0126-3 2 Set of 50 disposable mortar box.
CM 0126-4 Electronic console, with 4 measuring channels, and RS232 output interface for connecting up to 4 calorimeter bottles to the PC.